Digital Impact Testing Machine
We at TestCraft are Manufacturer, Supplier, Exporter of Analogue Impact Testing Machine, Digital Impact Testing Machine. Our setup is situated in Ichalkaranji, Maharashtra, India and majorly we serve customers from African and Gulf countries.
The pendulum is mounted on antifriction bearings. It has two starting positions, the upper one for Charpy the lower one for Izod testing. On release, the pendulum swings down to break the specimen & the energy absorbed in doing so is measured as the difference between the height of drop before rupture of the test specimen and is read from the maximum pointer position on the dial scale. (It is read on the digital readout in case of electronic machines). There are two strikers and one combined support anvil available for fitting into the pendulum and on the base of the machine for the Izod, Charpy test respectively. Changing from one striker to another is achieved simply by fixing the new striker into its position.
The Charpy test piece rests on alloy steel support anvils, fitted on the base of the machine rigidly held in position by Allen screws. An end stopper is provided for quickly and accurately locating the test piece centrally between the supports.
The Izod test piece is clamped vertically in Izod support fitted on the base of the machine. The support is provided with a machined vertical groove to suit the test piece size. The front clamp piece and the Allen screw enable clamping of the test piece incorrect height with the help of the Izod setting gauge supplied.